Did the Performance Verification software change when TLA Application Software v5.1 was introduced?
The Performance Verification software did change a little. The software for TLA7Lx/Mx/Nx/Px/Qx and TLA7Dx/Ex modules have not been updated to run on a remote PC controller. If you need to verify one of those modules, you must do so in a TLA7012 or TLA700-series mainframe. Also, the PV software will display a PCI-VXI bridge error when running with TLA Application Software version 5.x, unless you first delete the Visa32.dll file from the C:\Tekcats folder. Then when you see the message "UNABLE TO ENABLE SRQ EVENTS", you can continue by pressing the Enter key twice.
FAQ ID 55391
View all FAQs »