Is the interface cable from J103 on the 2711+ 2712+ 2714 or 2715 Spectrum Analyzer to J103 on the 2707 Tracking Generator a standard 1 to 1 configuration such as a VGA cable?
Mostly yes, but it is kept short (20 inches) and pin 10 is the center conductor of a small coaxial cable, with the shield at ground (pin 3), since there is a 26 MHz signal on that line. Order 012-1441-00 for the correct cable.
This FAQ Applies to:
No product series
FAQ ID 63796View all FAQs »