How do you set ACV DB function in Model DMM6500?
In the SCPI command set mode, the following commands will get you there:
This will set the unit to ACV, db and reference of 5.
If the unit is set to TSP, then use these commands:
dmm.measure.func = dmm.FUNC_AC_VOLTAGE
dmm.measure.unit = dmm.UNIT_DBM
dmm.measure.detectorbandwidth = dmm.DETECTBW_300HZ
This FAQ Applies to:
Product Series: Keithley DMM6500 6.5 Digit Multimeter with Graphical Touchscreen
FAQ ID 729491View all FAQs »