What is the difference between Position and Offset on an oscilloscope?
To answer this first you need to understand the layout of how your waveform trace is displayed.
On your oscilloscope you generally will see either 8 or 10 divisions vertically on the screen.
Along the left side of the screen you will see a marker which indicates your channels 0V level.
You can move the position of the channels 0 Voltage indicator vertically anywhere on the display.
If the signal you are probing has a voltage offset; you can compensate for the offset by applying a voltage in the scopes offset menu.
This FAQ Applies to:
Product Series: 5 Series B MSO 6 Series B MSO DPO7000 MSO / DPO70000 Mixed Signal Oscilloscope MSO/DPO2000B Mixed Signal Oscilloscope MSO3000 / DPO3000 MSO/DPO4000 Mixed Signal Oscilloscope (Discontinued) MSO/DPO5000B Mixed Signal Oscilloscope TDS2000C Digital Storage Oscilloscope (Discontinued) TDS3000C Digital Phosphor Oscilloscope (Discontinued)
FAQ ID 247211View all FAQs »