OM1106 Software Datasheet
Read Online:
The OM1106 Optical Modulation Analysis (OMA) software provides an ideal platform for research and testing of coherent optical systems. It is a complete software package for acquiring, demodulating, analyzing, and visualizing complex modulated systems from an easytouse interface. The software performs all calibration and processing functions to enable realtime burstmode constellation diagram display, eyediagram display, Poincaré sphere, and biterror detection. Advanced users can take advantage of the provided MATLAB signal analysis source code and modify the signal processing algorithms while still taking advantage of the rich user interface for acquisition, signal visualization, and numerical measurements.
Key features
 Complete coherent signal analysis system for polarizationmultiplexed QPSK, QAM, differential BPSK/QPSK, and other advanced modulation formats
 Displays constellation diagrams, phase eye diagrams, Qfactor, Qplot, spectral plots, Poincaré Sphere, signal vs. time, laser phase characteristics, BER, with additional plots and analyses available through the MATLAB interface ^{1}
 Measures polarization mode dispersion (PMD) of arbitrary order with most polarization multiplexed signals
 Smart polarization separation follows signal polarization
 User access to internal functions with a direct MATLAB interface
 Remote access available through Ethernet
 Superior user interface offers comprehensive visualization for easeofuse combined with the power of MATLAB
 Multicarrier software option allows userdefinable superchannel setup
 Superchannel configuration allows user to define number of channels, channel frequency, and channel modulation format
 Test automation acquires complete measurements at each channel
 Integrated measurement results allow easy channeltochannel comparisons
^{1} MATLAB is a registered trademark of The MathWorks, Inc.
Introduction
A common thread through the Tektronix OMseries coherent optical products is the OM1106 Optical Modulation Analysis (OMA) Software. This software is included with all Tektronix OM4000series optical modulation analyzers and is also available as a standalone software package for customers to use with their own optical analyzers or as a coherent optical research tool.
The OMA software consists of a number of major building blocks. At the heart of the software is a complete library of analysis algorithms. These algorithms have not been repurposed from the wireless communications world: they are specifically designed for coherent optical analysis, executed in a customersupplied MATLAB installation.
The OM1106 software also has a complete applications programmatic interface (API) to the application measurement and analysis algorithms, letting you perform detailed analysis of complex modulated optical signals without requiring any knowledge of MATLAB, analysis algorithms, or software programming.
Flexibility is a key strength of the OM1106 software. You can take measurements solely through the OMA software user interface, you can use the programmatic interface to and from MATLAB for customized processing, or you can do both by using the OMA software as a visualization and measurement framework around which you build your own custom processing.
OMA user interface
The OMA software has displays for all of the standard coherent optical visualizations such as eye diagrams, constellation diagrams, Poincaré spheres, and so on. The software also provides a complete measurement suite to numerically report key measurements. Constellation measurements include elongation, real and imaginary bias, magnitude, phase angle, EVM, and others. Eye diagram measurements include many key time domain metrics such as eye height, overshoot, undershoot, risetime, falltime, skew, crossing point, and so on. All numerical measurements also provide statistics to enable data gathering over longer periods of time.
OMA software showing colorgrade graphics options. Symbols can also be colored to a key indicating prior state. Data shown is 112 Gb/s PMQPSK.
OMA showing several equivalenttime measurement plots.
Get up and running fast with the easytouse OMA interface
The OMA software lets you easily configure and display your measurements and also provides software control for thirdparty applications using WCF or .NET communication. The software can also be controlled from MATLAB or LabVIEW.
The following image shows a QAM measurement setup. The plots can be moved, docked, or resized within the application work space or into separate windows on the desktop. You can display or close plots to show just the information you need.
QAM measurements on the OMA software.
In addition to the numerical measurements provided on individual plots, the Measurements tab provides a summary of all numeric measurements, including statistics.
The Measurements tab a summary of all measurements in one place
Make measurements faster
The OMA software is designed to collect data from the oscilloscope and move it into the MATLAB workspace with extreme speed to provide the maximum data refresh rate. The data is then processed in MATLAB to extract and display the resulting measurements.
Take control with tight MATLAB integration
Since 100% of the data processing occurs in MATLAB, test engineers can easily probe into the processing to understand each step along the way. R&D labs can also take advantage of the tight MATLAB integration by writing their own MATLAB algorithms for new techniques under development.
Use the optimum algorithm
Don’t worry about which algorithm to use. When you select a signal type in the OMA software (for example, PMQPSK), the application applies the optimal algorithm for that signal type to the acquired data. Each signal type has a specially designed signal processing approach optimized for that signal. This means that you get results right away.
Don’t get stymied by laser phase noise
Signal processing algorithms designed for electrical wireless signals don’t always work well with the much noisier sources used for complex optical modulation signals. Our robust signal processing methods tolerate enough phase noise to make it possible to test signals that would traditionally be measured by differential or direct detection such as DQPSK.
Find the right BER
Qplots are a great way to get a handle on your data signal quality. Numerous BER measurements vs. decision threshold are made on the signal after each data acquisition. Plotting BER vs. decision threshold shows the noise properties of the signal. Gaussian noise will produce a straight line on the Qplot. The optimum decision threshold and extrapolated BER are also calculated. This gives you two BER values: the actual counted errors divided by the number of bits counted, as well as the extrapolated BER for use when the BER is too low to measure quickly.
Qplot.
Constellation diagrams
Once the laser phase and frequency fluctuations are removed, the resulting electric field can be plotted in the complex plane. When only the values at the symbol centers are plotted, this is called a Constellation Diagram. When continuous traces are also shown in the complex plane, this is often called a Phase Diagram. Since the continuous traces can be turned on or off, we refer to both as the Constellation Diagram.
The scatter of the symbol points indicates how close the modulation is to ideal. The symbol points spread out due to additive noise, transmitter eye closure, or fiber impairments. The scatter can be measured by symbol standard deviation, error vector magnitude, or mask violations.
Constellation diagram.
Constellation measurements
Measurements made on constellation diagrams are available on the “flyout” panel associated with each graphic window. The measurements available for constellations are described below.
Measurement  Description 

Elongation  The ratio of the Q modulation amplitude to the I modulation amplitude is a measure of how well balanced the modulation is for the I and Q branches of a particular polarization’s signal 
Real Bias  Expressed as a percent, this says how much the constellation is shifted left or right. Real (Inphase) bias other than zero is usually a sign that the Inphase Tributary of the transmitter modulator is not being driven symmetrically at eye center 
Imag Bias  Expressed as a percent, this says how much the constellation is shifted up or down. Imaginary (Quadrature) bias other than zero is usually a sign that the Quadrature Tributary of the transmitter modulator is not being driven symmetrically at eye center 
Magnitude  The mean value of the magnitude of all symbols with units given on the plot. This can be used to find the relative sizes of the two Polarization Signals 
Phase Angle  The transmitter IQ phase bias. It should normally be 90 
StdDev by Quadrant  The standard deviation of symbol point distance from the mean symbol in units given on the plot. This is displayed for BPSK and QPSK 
EVM (%)  The RMS distance of each symbol point from the ideal symbol point divided by the magnitude of the ideal symbol expressed as a percent 
EVM Tab  The separate EVM tab shown in the right figure provides the EVM% by constellation group. The numbers are arranged to correspond to the symbol arrangement. This is ideal for setting Transmitter modulator bias. For example, if the left side groups have higher EVM than the right side, adjust the Inphase Transmitter modulator bias to drive the negative rail harder 
Mask Tab  The separate Mask tab shown in the right figure provides the number of mask violations by constellation group. The numbers are arranged to correspond to the symbol arrangement. The mask threshold is set in the Engine window and can be used for pass/fail transmitter testing 
Color features
The Color Grade feature provides an infinite persistence plot where the frequency of occurrence of a point on the plot is indicated by its color. This mode helps reveal patterns not readily apparent in monochrome. Note that the lower constellation groups of the example below have higher EVM than the top groups. In most cases this indicates that the quadrature modulator bias was too far toward the positive rail. This is not evident from the crossing points which are approximately correct. In this case an improperly biased modulator is concealing an improperly biased driver amp.
Color Grade Constellation.
Color Grade with fine traces.
Color Key Constellation Points is a special feature that works when not in Color Grade. In this case the symbol color is determined by the value of the previous symbol. If the prior symbol was in Quadrant 1 (upper right) then the current symbol is colored Yellow. If the prior symbol was in Quadrant 2 (upper left) then the current symbol is colored Magenta. If the prior symbol was in Quadrant 3 (lower left) then the current symbol is colored Light Blue (Cyan). If the prior symbol was in Quadrant 4 (lower right) then the current symbol is colored Solid Blue.
This helps reveal pattern dependence. The folowing figure shows that pattern dependence is to blame for the poor EVM on the other groups. The modulator nonlinearity would normally mask this type of pattern dependence due to RF cable loss, but here the improper modulator bias allows that to be transferred to the optical signal.
Color Key constellation
Eye diagrams
Eye diagram plots can be selected for appropriate modulation formats. Supported eye formats include Field Eye (the real part of the phase trace in the complex plane), Power Eye (which simulates the eye displayed with a Tektronix oscilloscope optical input), and DiffEye (which simulates the eye generated by using a 1bit delayline interferometer). As with the Constellation Plot you can rightclick to choose color options as well.
Field eye diagram.
Field eye measurements
Measurement  Description 

Q (dB)  Computed from 20 × Log10 of the linear decision threshold Qfactor of the eye 
Eye Height  The distance from the mean 1level to the mean 0level (units of plot) 
Rail0 Std Dev  The standard deviation of the 0level as determined from the decision threshold Qfactor measurement 
Rail1 Std Dev  The standard deviation of the 1level as determined from the decision threshold Qfactor measurement 
In the case of multilevel signals, the above measurements are listed in the order of the corresponding eye openings in the plot. The top row values correspond to the topmost eye opening.
The above functions involving Qfactor use the decision threshold method described in the paper by Bergano ^{1}. When the number of bit errors in the measurement interval is small, as is often the case, the Qfactor derived from the bit error rate may not be an accurate measure of the signal quality. However, the decision threshold Qfactor is accurate because it is based on all the signal values, not just those that cross a defined boundary.
^{1} N.S. Bergano, F.W. Kerfoot, C.R. Davidson, “Margin measurements in optical amplifier systems,” IEEE Phot. Tech. Lett., 5, no. 3, pp. 304306 (1993).
Additional measurements available for nonoffset formats
Measurement  Description 

Overshoot  The fractional overshoot of the signal. One value is reported for the tributary, and for a multilevel (QAM) signal it is the average of all the overshoots 
Undershoot  The fractional undershoot of the signal (overshoot of the negativegoing transition) 
Risetime  The 1090% rise time of the signal. One value is reported for the tributary, and for a multilevel (QAM) signal it is the average of all the rise times 
Falltime  The 9010% fall time of the signal 
Skew  The time relative to the center of the power eye of the midpoint between the crossing points for a particular tributary 
Crossing Point  The fractional vertical position at the crossing of the rising and falling edges 
Measurements vs. Time
In addition to the eye diagram, it is often important to view signals versus time. For example, it is instructive to see what the field values were doing in the vicinity of a bit error. All of the plots that display symbolcenter values will indicate if that symbol errors by coloring the point red (assuming that the data is synchronized to the indicated pattern). The Measurement vs. Time plot is particularly useful to distinguish errors due to noise, pattern dependence, or pattern errors.
Errored symbol (red dot) in Measurement vs. Time plot.
3D visualization tools
Complexmodulation signals are inherently 3D since inphase and quadrature components are being changed vs. time. The 3D Eye Diagram provides a helpful combination of the Constellation and Eye diagrams into a single 3D diagram. This helps to visualize how the complex quantity is changing through the bit period. The diagram can be rotated and scaled.
Also available in 3D is the Poincaré Sphere. The 3D view is helpful when viewing the polarization state of every symbol. The symbols tend to form clusters on the Poincaré Sphere which can be revealing to expert users. The nonnormalized Stokes Vectors can also be plotted in this view.
Analysis Controls
The various Analysis Control tabs in the OMA software let you set parameters for the system and relevant measurements.
MATLAB tab.
Signal Spectrum tab.
Parameter  Description 

Frequency  Clock recovery is performed in software, so only a frequency range of expected clock frequencies is required 
Signal Type  The signal type (such as PMQPSK) determines the algorithm used to process the data 
Data Patterns  Specifying the known PRBS or user pattern by physical tributary permits error counting, constellation orientation, and twostage phase estimation 
You can assign user data patterns in the MATLAB tab. The data pattern can be input into MATLAB or found directly through measurement of a high SNR signal.
Signal spectra
An FFT of the corrected electric field vs. time can reveal much about the data signal. Asymmetric or shifted spectra can indicate excessive laser frequency error. Periodicity in the spectrum shows correlation between data tributaries. The FFT of the laser phase vs. time data can be used to measure laser phase noise.
Laser Phase Spectrum window
Poincaré Sphere
Polarization data signals typically start out well aligned to the PMfiber axes. However, once in standard single mode fiber, the polarization states will start to drift. However, it is still possible to measure the polarization states and determine the polarization extinction ratio. The software locks on each polarization signal. The polarization states of the two signals are displayed on a circular plot representing one face of the Poincaré sphere. States on the back side are indicated by coloring the marker blue. The degree of orthogonality can be visualized by inverting the rear face so that orthogonal signals always appear in the same location with different color. So, Blue means back side (negative value for that component of the Stokes vector), X means Xtributary, O means Ytributary, and the Stokes vector is plotted so that left, down, blue are all negative on the sphere.
InvertedRearFace – Checking this box inverts the rear face of the Poincaré sphere display so that two orthogonal polarizations will always be on top of each other.
Poincaré Sphere window.
Impairment measurement and compensation
When studying transmission implementations, it is important to be able to compensate for the impairments created by long fiber runs or optical components. Chromatic Dispersion (CD), and Polarization Mode Dispersion (PMD) are two important linear impairments that can be measured or corrected by the OMA software. PMD measurement is based on comparison of the received signal to the backtoback transmitter signal or to an ideal signal. This produces a direct measure of the PMD instead of estimating based on adaptive filter behavior. You can specify the number of PMD orders to calculate. Accuracy for 1storder PMD is ~1 ps at 10 Gbaud. There is no intrinsic limit to the CD compensation algorithm. It has been used successfully to compensate for many thousands of ps/nm.
Recording and playback
You can record the workspace as a sequence of .MAT files using the Record button in the Offline ribbon. These files are recorded in a default directory, usually the MATLAB working directory, unless previously changed. You can play back the workspace from a sequence of .MAT files by first using the Load button in the Offline Commands section of the Home ribbon. Load a sequence by marking the files you want to load using the Ctrl key and marking the filenames with the mouse. You can also load a contiguous series using the Shift key and marking the first and last filenames in the series with the mouse. Use the Run button in the Offline Commands section of the Home ribbon to cycle through the .MAT files you recorded. All filtering and processing you have implemented occurs on the recorded files as they are replayed.
Workspace record and playback.
Multicarrier superchannel support
Even as 100G coherent optical systems are being deployed, architectures for 400G and beyond are being proposed and developed. One architecture gaining prominence is the “superchannel.” The configurations of superchannels vary considerably. Some proposals call for 400G to be achieved by 2 carriers of DP16QAM. Other proposals are for 500 Gb/s consisting of 10 or more carriers of DPQPSK. Some of these carriers are arranged on a standard ITU carrier grid, while others support 12.5 GHz “gridless” layouts. Clearly, flexible test tools are needed for such nextgeneration systems. Option MCS for the OM4245 or OM4225 offers the complete flexibility to carry out such tests.
Multicarrier setup.
Multicarrier measurements.
Userdefinable superchannels
For manufacturers getting a jump on superchannels, or researchers investigating alternatives, userdefinable superchannel configurations are a must. Option MCS lets you set up as many carriers within the superchannel definition as necessary. Each carrier can have an arbitrary center frequency; no carrier grid spacing is imposed. The carrier center frequencies can be set as absolute values (in THz) or as relative values (in GHz).
Typically, the application software retunes the OM4245 local oscillator for each carrier. However, in cases where multiple carriers will fit within the oscilloscope bandwidth, multiple carriers can be demodulated in software from a common local oscillator frequency. This provides the flexibility to specify the preferred local oscillator frequency for each carrier.
Superchannel spectrum.
Automated measurements
Once the superchannel is configured, the OMA software can take measurements on each channel without further intervention. The software automatically tunes the OM4245 local oscillator, takes measurements at that channel, retunes to the next channel, and so on, until measurements of the entire superchannel have been taken. Results of each channel are displayed in realtime and persist after all measurements are made for easy comparison.
Integrated measurement results
All of the same measurement results that are made for single channels are also available for individual channels in a superchannel configuration. Additionally, multicarrier measurement results are available sidebyside for comparison between channels. Visualizations such as eye diagrams, constellation diagrams, and optical spectrum plots can be viewed a single channel at a time, or with all channels superimposed for fast comparison.
For separating channels in a multicarrier group, several different filters can be applied, including raised cosine, Bessel, Butterworth, Nyquist, and userdefined filters. These filters can be any order or rolloff factor and track the signal frequency.
Coherent optical signal generation
Tektronix offers several signal generation instruments capable of generating coherent optical waveforms. The AWG70000 Series Arbitrary Waveform Generators (AWG) and the PPG3000 Series Programmable Pattern Generators offer the flexibility to choose the signal generator best suited to your test requirements.
The AWG70000 Series can reach sampling rates as high as 50 GS/s with 10 bits vertical resolution. This level of performance allows for the direct generation of IQ baseband signals required by modern coherent optical communication systems. The arbitrary waveform generation capabilities of the AWG70000 Series makes it possible to create multilevel signals such as 16QAM or 64QAM, add impairments to a signal, or create waveforms that are precompensated for the realworld effects of the test system.
Coherent optical signal generation is one of the more demanding applications for an AWG. The requirements in terms of number of channels, sampling rate, bandwidth, record length, and timing and synchronization quality can be only met by the highest performance instruments, such as the Tektronix AWG70000 series. The unique capability of generating ideal or distorted signals, and the ease to add new modulation schemes and signal processing algorithms without the need to add any extra hardware, make Tektronix AWGs an ideal tool for coherent optical communication research and development.
The PPG3000 Series can generate patterns up to 32 Gb/s and offers 1, 2, or 4 channels in a single instrument. The patterns can be standard PRBS patterns or userdefined. Using a 4channel pattern generator makes creating dualpolarization IQ waveforms very simple.
Receiver test
The OMA software can take receiver test measurements as specified in the OIF Implementation Agreement for Integrated Dual Polarization Intradyne Coherent Receivers. These measurements include:
 Symbol rate
 Operating signal power
 Local oscillator power
 Linear output swing adjustment range
 Maximum gain control bandwidth
 Total harmonic distortion
 Common mode rejection
 Small signal bandwidth
 Low frequency cutoff
 Optical reflectance
 Electrical return loss
 Skew (P, N)
 Channel skew
 Channel skew variation
Relevant measurements are a function of wavelength. In order to optimize test time, you can specify the start and stop wavelengths, the step size, and a wide range of other parameters. The Wavelength Sweep can also produce a calibration file that is used by software to correct for coherent receiver impairments, when the receiver is part of a Tektronix Optical Modulation Analyzer system.
Test results are displayed both graphically and numerically based upon the user's preference. Some receiver test measurements require the use of the OM2210 Receiver Calibration Source.
Interaction between OMA and MATLAB
The OMA software takes information about the signal provided by the user together with acquisition data from the oscilloscope and passes them to the MATLAB workspace. A series of MATLAB scripts are then called to process the data and produce the resulting field variables. The software then retrieves these variables and plots them. Automated tests can be accomplished by connecting to the OMA software or by connecting directly to the MATLAB workspace.
You do not need any familiarity with MATLAB; the OMA software can manage all MATLAB interactions. However, advanced users can access the MATLAB interface internal functions to create userdefined demodulators and algorithms, or for custom analysis visualization.
OMA software/MATLAB data flow
Signal processing approach
For realtime sampled systems, the first step after data acquisition is to recover the clock and retime the data at 1 sample per symbol at the symbol center for the polarization separation and following algorithms (shown as upper path in the figure). The data is also resampled at 10X the baud rate (user settable) to define the traces that interconnect the symbols in the eye diagram or constellation (shown as the lower path).
The clock recovery approach depends on the specified signal type. Laser phase is then recovered based on the symbolcenter samples. Once the laser phase is recovered, the modulation part of the field is available for alignment to the expected data for each tributary. At this point bit errors can be counted by looking for the difference between the actual and expected data after accounting for all possible ambiguities in data polarity. The software selects the polarity with the lowest BER. Once the actual data is known, a second phase estimate can be done to remove errors that may result from a laser phase jump. Once the field variables are calculated, they are available for retrieval and display by the OMA software.
At each step the best algorithms are chosen for the specified data type, requiring no user intervention unless desired.
Data flow through the “Core Processing” engine.
Signal processing customization
The OM1106 software includes the MATLAB source code for the "CoreProcessing" engine (certain proprietary functions are provided as compiled code). You can customize the signal processing flow, or insert or remove processes as desired. Alternatively, you can remove all Tektronix processing and completely replace it with your own. By using the existing variables defined for the data structures, you can then see the results of analysis processing using the rich visualizations provided by the OMA software. This lets you focus your time on algorithms rather than on tasks such as acquiring data from the oscilloscope or displaying constellation diagrams.
Dynamic MATLAB integration
Customizing the CoreProcessing algorithms provides an excellent way to conduct signal processing research. In order to speed up development of signal processing the OM1106 software provides a dynamic Matlab integration window. Any Matlab code typed in this window is executed on every pass through the signal processing loop. This lets you quickly add or "comment out" function calls, write specific values into data structures, or modify signal processing parameters on the fly, without needing to stop the processing loop or modify the Matlab source code.
CoreProcessing functions
The following are some of the CoreProcessing functions used to analyze the coherent signal. Full details on these functions, their use in the processing flow, and the MATLAB variable used, are available in the OM1106 user manual.
 EstimateClock determines the symbol clock frequency of a digital datacarrying optical signal based on oscilloscope waveform records. The scope sampling rate may be arbitrary (having no integer relationship) compared to the symbol rate.
 ClockRetime forms an output parameter p, representing a dualpolarization signal vs. time, from four oscilloscope waveforms V. The output p is retimed to be aligned with the timing grid specified by Clock.
 EstimateSOP reports the state of polarization (SOP) of the tributaries in the optical signal. The result is provided in the form of an orthogonal (rotation) matrix RotM. For a polarization multiplexed signal the first column of RotM is the SOP of the first tributary, and the second column the SOP of the second tributary. For a single tributary signal, the first column is the SOP of the tributary, and the second column is orthogonal to it. The signal is transformed into its basis set (the tributaries horizontal vertical polarizations) by multiplying by the inverse of RotM.
 EstimatePhase estimates the phase of the optical signal. The algorithm used is known to be close to the optimal estimate of the phase. The algorithm first determines the heterodyne frequency offset and then estimates the phase. The phase reported in the .Values field is after the frequency offset has been subtracted.
 ApplyPhase multiplies the values representing a single or dual polarization parameters vs. time by a phase factor to give a resulting set of values.
 AlignTribs performs ambiguity resolution. The function acts on variable which is already corrected for phase and state of polarization, but for which the tributaries have not been ordered. AlignTribs uses the data content of the tributaries to distinguish between them. AlignTribs processes the data patterns in order according to the modulation format, starting with XI. For each pattern it tries to match the given data pattern with the available tributaries of the signal. If the same pattern is used for more than one tributary, the relative pattern delays will be used to distinguish between them.
The use of delay as a secondary condition to distinguish between tributaries means that AlignTribs will work with transmission experiments that use a single data pattern generator which is split several ways with different delays. The delay search is performed only over a limited range of 1000 bits in the case of PRBS patterns, so this method of distinguishing tributaries will not usually work when using separate data pattern generators programmed with the same PRBS.
 Spect estimates the power spectral density of the optical signal using a discrete Fourier transform. It can take any many of our time waveforms as input such as corrected oscilloscope input data, frontend processed data, polarization separated data, averaged data, and FIR data. It can also apply Hanning or FlatTop window filters and produce the desired resolution bandwidth over a set frequency range.
 GenPattern generates a sequence of logical values, 0s and 1s, given an exact data pattern. The exact pattern specifies not only the form of the sequence but also the place it starts and the data polarity. The data pattern specified may be a pseudorandom bit sequence (PRBS) or a specified sequence.
 LaserSpectrum estimates the power spectral density of the laser waveform in units of dBc. The function LaserSpectrum takes ThetaSym, the estimated relative laser phase sampled at the symbol rate, as input and defines the frequency centered laser waveform. This waveform is then scaled by a hamming window, and the power spectral density of the waveform is estimated as the discrete Fourier transform of this signal.

QDecTh uses the decision threshold method to estimate the Qfactor of a component of the optical signal. The method is useful because it quickly gives an accurate estimate of Qfactor (the output signaltonoise ratio) even if there are no bit errors, or if it would take a long time to wait for a sufficient number of bit errors.
Supported measurements and display tools
Characteristic  Description 
Realtime supported feature 
Equivalent time supported feature 

Constellation Diagram  Constellation diagram accuracy including intradyne and demodulation error can be measured by the RMS error of the constellation points divided by the magnitude of the electric field for each polarization signal  X  X 
Constellation Elongation  Ratio of constellation height to width  X  X 
Constellation Phase Angle  Measure of transmitter IQ phase angle  X  X 
Constellation I and Q Bias  Measure of average symbol position relative to the origin  X  X 
Constellation Mask  Usersettable allowed EVM level. Symbols violating the mask are counted  X  X 
Eye Decision Threshold Qfactor  The actual Q achieved will depend on the quality of the data signal, the signal amplitude, and the oscilloscope used for digitalization. Using the Tektronix DPO73304DX oscilloscope (4Ch), a Qfactor of 20 dB is achievable at 40 GBaud  X  X 
Decision Threshold Qplot  Displays BER vs. decision threshold for each eye opening. The Q value at optimum decision threshold is the Qfactor  X  X 
Signal Spectrum and Laser Spectrum  Display of signal electric field vs. time in the complex plane FFT of power signal or laser phase noise  X  
MATLAB Window  Commands may be entered that execute each time signals are acquired and processed  X  X 
Measurements vs. Time  Predefined measurement vs. time measurements include Optical field, symbolcenter values, errors, and averaged waveforms. Any parameter can be plotted vs. time using a customcreated MATLAB expression  X  X 
3D Measurements  3D Eye (complex field values vs. time), and 3D Poincaré Sphere for symbol and tributary polarization display  X  
Differential Eye Diagram Display  Balanced or singleended balanced detection is emulated and displayed in the Differential Eye Diagram  X  
Frequency Offset  Frequency offset between signal and reference lasers is displayed in Measurement panel  X  
Poincarè Sphere  Polarizations of the Polmuxed signal tributaries are tracked and displayed on the Poincaré Sphere. PER is measured  X  X 
Signal Quality  EVM, Qfactor, and mask violations  X  X 
Tributary Skew  A time offset for each tributary is reported in the Measurement panel  X  X 
CD Compensation  No intrinsic limit for offline processing – FIRbased filter to remove CD in frequency domain based on a given dispersion value  X  
PMD Measurement  PMD values are displayed in the Measurement panel for Polarizationmultiplexed formats with a userspecified number of PMD orders  X  
Oscilloscope and/or Cable Delay Compensation  The OMA software corrects cable, oscilloscope, and receiver skew through interpolation. Additional cable adjustment is available using the oscilloscope UI  ±0.5 ns  
Oscilloscope Skew Adjustment  Equivalenttime oscilloscope skew is adjusted using the "Delay" feature in the supported sampling head plugins  ±100%  
Calibration Routines  Receiver Skew, DC Offset, and Path Gain Mismatch Hybrid angle and state of polarization are factory calibrated  X  X 
Data Export Formats  MATLAB (other formats available through MATLAB or ATE interface); PNG  X  X 
Raw Data Replay with Different Parameter Setting  Movie mode and reprocessing  X  X 
Bit Error Ratio Measurements  Number of counted bits/symbols  X  X 
Number or errors detected  X  X  
Bit error ratio  X  X  
Differentialdetection errors  X  X  
Save acquisition on detected error  X  X  
Offline Processing  Run software on a separate PC or on the oscilloscope  X  X 
Coherent Eye Diagram  Shows the InPhase or Quadrature components vs. time modulo two bit periods.  X  X 
Power Eye Diagram  Shows the computed power per polarization vs time modulo 2 bit periods.  X  X 
System requirements
Supported platforms for the OM1106 Optical Modulation Analysis (OMA) software:
 Computer with nVidia graphics card running US Windows 7 64bit and MATLAB 2011b (64bit)
 Computer with nVidia graphics card running US Windows XP 32bit and MATLAB 2009a (32bit)
The following platforms are supported but may not be able to use certain advanced graphics features such as color grade and 3D:
 Tektronix 70000 Series Oscilloscopes running Windows 7 64bit and MATLAB 2011b (64bit)
 Computer with nonnVidia graphics running US Windows 7 64bit and MATLAB 2011b (64bit)
 Computer with nonnVidia graphics running US Windows XP 32bit and MATLAB 2009a (32bit)
Please check with Tektronix when ordering for the most uptodate requirements including support for the latest releases of MATLAB software.
Please contact Tektronix for a price quote or to arrange a demonstration. All product descriptions and specifications are subject to change without notice.
85W310372