What is the maximum bus width that can be monitored?
You can combine modules using the special high-speed connectors on the side of each logic analyzer module so that all channels operate from a single timebase. This is called "module merging." All channels may be contained within a single channel group (useful for triggering or data display) or divided among several channel groups.
The maximum number of merged channels for TLA7Lx/Mx/Nx/Px/Qx modules is three 136 channel modules for a total of 408 channels.
The maximum number of merged channels for TLA7AAx/ABx/NAx/BBx modules is five 136 channel modules for a total of 680 channels.
常見問答集 ID 61266檢視所有常見問答集 »