Most technicians have been taught to acquire a signal that is full screen to make the most accurate measurements. What can prevent clipping of signals that exceed the display range?
Dynamic range is stated in divisions. HP and LeCroy use 8 divisions of dynamic range. Any waveform exceeding the display will be clipped. Tektronix TDS oscilloscopes have 10.24 divisions of dynamic range, more than one division above the display top and below the bottom. This helps limit the opportunity for clipping. A 0.5 zoom factor will recover these divisions to the display for viewing.
Diese FAQs beziehen sich auf:
Alle FAQs ansehen »