연락처

텍트로닉스 담당자와 실시간 상담 6:00am-4:30pm PST에 이용 가능

전화

전화 문의

9:00am-6:00PM KST에 이용 가능

다운로드

매뉴얼, 데이터 시트, 소프트웨어 등을 다운로드할 수 있습니다.

다운로드 유형
모델 또는 키워드

피드백

# Specs: How do I apply the temperature coefficient specification?

### 질문:

Specs: How do I apply the temperature coefficient specification?

### 답변:

What is Temperature Coefficient and how is it applied?

Temperature Coefficient is the specification that is added to the general accuracy specification when the measuring instrument is at an ambient temperature that is outside the normal operating temperature.

It is calculated by the following manner:

The specification is normally defined as percent per degree. This means for each degree outside of the normal operating temperature this percentage is added to the normal accuracy specification. In general, the Temperature Coefficient is much less than the normal accuracy. The normal operating temperature is 18 to 28 oC.

Example:

Temperature Coefficient = 0.0001%/C.

Accuracy = 20ppm of reading + 20ppm of range.

The accuracy in measuring a 0.5V signal on the 1V range for the 90 day specification is 30ppm or 0.03% or 30 microvolts. (Refer to accuracy calculation FAQ for more details.)

Now consider that the unit is at 33 oC ambient temperature. Since this is five oC outside the normal operating temperature, the Temperature Coefficient is going to apply to the normal specification.

The Temperature Coefficient is stated as 0.0001% for each oC

0.0001% = 0.000001 = 1 ppm

5 x 1ppm = 5ppm (or 0.0005%)

5ppm added to the normal specification of 30ppm is 35 ppm or 0.00035% total uncertainty. For the 0.5 volt signal being measured, this results in 35 microvolts of uncertainty.

A reported reading in the range of 0.499965 to 0.500035 will be within tolerance.

이 FAQ는 다음에 적용 됩니다:

제품 시리즈 없음

제품:

FAQ ID 71941

모든 FAQ 보기 »