When I plug my one of these electrical sampling modules (80E07+ 80E08+ 80E09+ 80E10) into my 8000-series oscilloscope it locks up. What do I need to do?
The 8000-series sampling oscilloscope must have the following in order to be compatible with the 80E07, 80E08, 80E09, and 80E10:
- Win2K operating system
- Product SW (firmware) version 220.127.116.11, or higher
The latest product SW can be downloaded from our website: www.tektronix.com, under SW and Drivers.
이 FAQ는 다음에 적용 됩니다:
제품 시리즈 없음
FAQ ID 64276모든 FAQ 보기 »