연락처

텍트로닉스 담당자와 실시간 상담 6:00am-4:30pm PST에 이용 가능

전화

전화 문의

9:00am-6:00PM KST에 이용 가능

다운로드

매뉴얼, 데이터 시트, 소프트웨어 등을 다운로드할 수 있습니다.

다운로드 유형
모델 또는 키워드

피드백

# What is the effective input resistance on the 6517B electrometer in current mode?

### 질문:

What is the effective input resistance on the 6517B electrometer in current mode?

### 답변:

Voltage burden is what is important here. Model 6517B is as close to an ideal ammeter as its voltage burden is extremely low.
However, you can calculate the "effective" input resistance of each range.
20pA range is 20uV / 20pA = 1M ohm
200pA range is 100uV / 200pA = 500k ohm
2nA range is 20uV / 2nA = 10k ohm
20nA range is 20uV / 20nA = 1k ohm
200nA range is 100uV / 200nA = 500 ohm
2uA range is 20uV / 2uA = 10 ohm
20uA range is 20uV / 20uA = 1 ohm
200uA range is 100uV / 200uA = 0.5 ohm
2mA range is 2mV / 2mA = 1 ohm
20mA range is 5mV / 20mA = 0.2 ohms
This is effective input resistance.
The feedback - is the closest you can get to an ideal ammeter.
This means the voltage burden is almost zero. Look in the specs for each range.
So the effective input resistance is just a number. It usually does not matter.

이 FAQ는 다음에 적용 됩니다:

제품:

FAQ ID 774026

모든 FAQ 보기 »