What is the difference between the P7360 and P7380 Non A and the new P7360A and P7380A probes?
The difference between the standard probe and the "A" version is that the "A" version will provide the customer greater Dynamic range in the 5X Attenuation range. The changes to the specifications are shown below.
P7360 and P7380: Dynamic Range is specified at 0.625V (Total 1.25 volts Peak-to-Peak) in the X5 attenuation range.
P7360A and P7380A: Dynamic Range is specified at 1V (2.0 volts Peak-to-Peak) in the X5 attenuation range.
This FAQ Applies to:
No product series
FAQ ID 54921View all FAQs »