The Tektronix M-PHY TX Essentials Test software runs on Tektronix real-time oscilloscopes that are based on Windows 7 computer operating systems. M-PHY TX Essentials provides support for all the base measurements for Spec 3.1 & CTS 3.1 within DPOJET framework, a state of the art tool designed for jitter analysis and core measurements. This solution is designed for engineers doing characterization and margin testing of the device. The solution is supplemented with MOI which provides guidance to users to setup the test and scope to accomplish specific measurements.
The Tektronix M-PHY TX Automated Test software runs on Tektronix real-time oscilloscopes that are based on Windows 7 computer operating systems. M-PHY TX Automated provides support for 100% of tests as per Spec 3.1 & CTS 3.1 using TekExpress 4.0 framework, a state of the art tool designed for automation. The backend engine of automation is Iron Python based which uses socket based programming. This solution is designed for engineers doing verification and validation as per the CTS.
Option M-PHY Essentials enables full customization and comprehensive characterization, debug and analysis using setup libraries and MOI in DPOJET; the state of the art jitter analysis and debugging tool
Full Contour Extrapolated eye diagram for BER analysis in steps at E-6 to E-10
The Physical layer is implemented for various high-speed serial interfaces (chip to chip communication) inside mobile devices. The common interfaces which utilize the M-PHY solution are:
Companion chip interface with application processor over Unipro/SSIC
Once the test bench is set up and the DUT is properly connected, simply press the Run button to perform the selected test suite.
Single-button Automated M-PHY Transmitter testing
You do not need to be an expert on testing procedures. Remembering the exact steps to take each measurement is time consuming and often requires going back to the M-PHY specifications. M-PHY TX takes the guesswork out of conducting M-PHY Transmitter testing.
Even if you remember how to use the test equipment, it is common for even the most experienced operators to forget steps in the procedure or to set up the correct parameters, such as applying the correct signal impairments for a given test. M-PHY TX allows engineers to simply select the desired tests to run, and then work on other tasks while the tests are being executed.
Test setup and test execution is very simple with the M-PHY TX Automated software. The test setup connections are very minimal, as it involves only one-piece of equipment for M-PHY TX. The TekExpress software provides a Graphical User Interface (GUI) and an intuitive workflow through setup and testing.
When setting up a test, nothing can be simpler than hooking up the test system by looking at a schematic. View the schematic of the selected test with a click of a button.
The M-PHY TX Report tab provides a single printable report of all the tests along with a Pass/Fail summary table, margins, and optionally waveform screen captures, eye diagrams, histograms, bathtub charts, etc.
DPOJET software with Option M-PHY provides the essential set of M-PHY Transmitter measurements with greater flexibility in the test setup. Like D-PHY Essentials on DPOJET, M-PHY Essentials also enables Characterization, Debug, Analysis, and Conformance testing of M-PHY designs.
M-PHY Transmitter testing with M-PHY TX using Single-ended/Differential probes
M-PHY Essentials supports complete measurements to be performed in High Speed mode. It includes unique measurements such as Power Spectral Density on the real-time oscilloscope itself, which is a patent-pending methodology supported by Tektronix. Other solutions in the market require additional hardware to perform this test.
M-PHY Essentials slew rate measurement can be extended to slew rate resolution measurement.
Transmitter Eye diagram measurement for HS G1 with Mask Hits
Transmitter Eye diagram measurement for HS G1 Auto shape to adjust mask hits and optimal eye opening
For M-PHY HS G3, the mask is defined like a diamond shape as shown below. For M-PHY Eye diagram prorated parameter values for a BER of 1E-6.
Transmitter Eye diagram measurement for HS G3 with BER contour
For HS G3B the eye diagram shows the accumulated eye at BER E-6 and extrapolated eye at various values in contour format. The extrapolated eye is at BER of 1E-10. Tektronix provides the unique ability for extrapolation using contours. This provides the designers the level of confidence of their design, a good perspective on margin for their designs and allows them to accomplish this task in less time without having to accumulate the eye at BER-10.
Transmitter Eye diagram measurement for HS G3B
For M-PHY TX testing, measurements are specified at the TX pins. Many a times, the users measure signals at the end of the channel to view the effect at the RX pins or some test point in the middle of the channel (Mid-bus probing). For conformance testing, to ensure that the values meet the CTS, there would have a need to embed and/or de-embed channel to make measurements at the pin.
TekExpress Automated software allows you to do mid-bus probing via embedding/de-embedding signal path using filter files. You can use the 'Filter Setup' option in DUT panel to add the filters. This setting will be applied globally to all the measurements during acquisition.
The requirements for measurement equipment used to test conformance of an M-PHY transmitter running in High Speed mode is summarized below.
|Return Loss||Per specification limits|
|Differential termination||100Ω across input|
|Common mode termination||Infinite or consistent value|
|Rise Time||3X fastest signal rise time|
|Sensitivity||200-300 mVFS capability|
|Noise||Minimal added noise, (<1 or 2 mVrms desirable)|
|Attenuation||Smallest attenuation possible (1X desirable)|
The MIPI M-PHY standard presents significant challenges for oscilloscopes and probing. These challenges result in stringent requirements for HS-MODE measurements. SMA style probes with 50Ω inputs have been shown to yield superior results compared to high impedance probe approaches, particularly for HS-GEAR3 speeds. M-PHY CTS has been updated to list support for “SMA probe” type probing solutions. Tektronix SMA probes like the P7633 are unique in that and they can meet the requirements of M-PHY testing and introduce lower noise than the other oscilloscope probes.
Tektronix P7633 SMA probe
For more details on probe specifications, refer to the probes datasheet:
For a complete list of required equipment, visit http://www.tek.com/MIPI.
1.1.1 – HS-TX Unit Interval and Frequency Offset (UIHS and fOFFSET-TX)
1.1.2 – HS-TX Common-Mode AC Power Spectral Magnitude Limit (PSDCM-TX) (Informative only)
1.1.3 – HS-TX PREPARE Length (THS-PREPARE)
1.1.4 – HS-TX Common-Mode DC Output Voltage Amplitude (VCM-TX)
1.1.5 – HS-TX Differential DC Output Voltage Amplitude (VDIF-DC-TX)
1.1.6 – HS-TX G1 and G2 Differential AC Eye (TEYE-TX, VDIF-AC-TX)
1.1.7 – HS-TX G3 Differential AC Eye (TEYE-HS-G3-TX, VDIF-AC-HS-G3-TX)
1.1.8 – HS-TX 20/80% Rise and Fall Times (TR-HS-TX and TF-HS-TX)
1.1.9 – HS-TX Lane-to-Lane Skew (TL2L-SKEW-HS-TX)
1.1.10 – HS-TX Slew Rate Control Range (SRDIF-TX[MAX/MIN])
1.1.11 – HS-TX Slew Rate State Monotonicity
1.1.12 – HS-TX Slew Rate State Resolution (ΔSRDIF-TX)
1.1.13 – HS-TX Intra-Lane Output Skew (TINTRA-SKEW-TX)
1.1.14 – HS-TX Transmitter Pulse Width (TPULSE-TX)
1.1.15 – HS-TX Total Jitter (TJTX)
1.1.16 – HS-TX Short-Term Total Jitter (STTJTX)
1.1.17 – HS-TX Deterministic Jitter (DJTX)
1.1.18 – HS-TX Short-Term Deterministic Jitter (STDJTX)
BER Contour for extrapolated eye capability available for Test 1.1.6 and Test 1.1.7 with Scope Option DJAN
1.2.1 – PWM-TX Transmit Bit Duration (TPWM-TX)
1.2.2 – PWM-TX Transmit Ratio (kPWM-TX)
1.2.3 – PWM-TX PREPARE Length (TPWM-PREPARE)
1.2.4 – PWM-TX Common Mode DC Output Voltage Amplitude (VCM-TX)
1.2.5 – PWM-TX Differential DC Output Voltage Amplitude (VDIF-DC-TX)
1.2.7 - PWM-TX Maximum Differential AC Output Voltage Amplitude (VDIF-AC-TX)
1.2.8 – PWM-TX 20/80% Rise and Fall Times (TR-PWM-TX and TF-PWM-TX)
1.2.9 – PWM-TX Lane-to-Lane Skew (TL2L-SKEW-PWM-TX)
1.2.10 – PWM-TX G1 Transmit Bit Duration Tolerance (TOLPWM-G1-TX)
1.2.11 – PWM-TX G0 Minor Duration (TPWM-MINOR-GO-TX)
1.3.1 – SYS-TX Unit Interval and Frequency Offset (UISYS and fOFFSET-TX)
1.3.2 – SYS-TX RefClk Frequency (UIREFCLK and fREFCLK-TX)
1.3.3 – SYS-TX PREPARE Length (TSYS-PREPARE)
1.3.4 – SYS-TX Common Mode DC Output Voltage Amplitude (VCM-TX)
1.3.5 – SYS-TX-Differential DC Output Voltage Amplitude test (VDIF-DC-TX)
1.3.7 – SYS-TX Maximum Differential AC Output Voltage Amplitude (VDIF-AC-TX)
1.3.8 – SYS-TX 20/80% Rise and Fall Times (TR-SYS-TX and TF-SYS-TX)
1.3.9 – SYS-TX Lane-to-Lane Skew (TL2L-SKEW-SYS-TX)
|DPO/DSA/MSO70000C/DX||DPO (Digital Phosphor Oscilloscope), DSA (Digital Serial Analyzer), or MSO (Mixed Signal Oscilloscope) Oscilloscopes with Option DJA and Option DJAN (for BER Contours). Option DJA is mandatory. Option DJAN is optional but required for BER Contours.|
The following bandwidths are needed:
|DPO/DSA/MSO70000C/DX Option M-PHY TX 1 |
DPO-UP Option M-PHY TX
|M-PHY Automated Transmitter Solution|
|DPOFL-M-PHY TX||M-PHY Automated Transmitter Solution (Floating License version)|
|DPO/DSA/MSO70000C/DX Option M-PHY|
DPO/MSO70000CGSA Option M-PHY
DPO-UP/DPO7UP Option M-PHY
|MIPI® M-PHY Essentials|
|DPOFL-M-PHY||MIPI® M-PHY Essentials (Floating License version)|
1 Requires DPOJET Jitter and Eye Analysis Tools (Option DJA is mandatory and Option DJAN is optional but required for BER Contours).
|PGY-UPRO 1||M-PHY UniPro Protocol Decode (3rd-party software)|
|PGY-LLI 1||M-PHY LLI Protocol Decode (3rd-party software)|
|PGY-DGRF 1||M-PHY DigRFv4 CommView Protocol Decode (3rd-party software)|
|DPO-UP Option SR-810B||8b/10b Serial Analysis|
|PGY-SSIC 1||M-PHY SSIC Decode (3rd-party software)|
For more details on the Protocol Decoders, refer to the Protocol Decode Datasheet at http://www.tek.com/MIPI.
1 Requires Option ST6G Serial Protocol Triggering.
|Gear type||Data rate||Fixtured / RF connection|
|HS-Gear1 1||1.46 Gb/s||P7313SMA or P7625 or P7633|
|HS-Gear2||2.92 Gb/s||P7313SMA or P7625 or P7633|
|HS-Gear3||5.83 Gb/s||P7625 or P7633|
|PWM Gears (G0 to G7) 2||10 Kbps to 576 Mbps||P7313|
|SYS Gears||26 MHz|
38.4 M Hz
52 M Hz
PGY-UPRO and PGY-LLI require differential probing. For the DPO70000 Series, P7600 and P7300 Series probes are well suited.
Note: In Differential mode acquisition of M-PHY TX automated software (i.e. when using only one differential probe per lane), Single-ended measurements cannot be performed. However, while using Differential mode for acquisitions users can connect up to 4 lanes of a M-PHY transmitter DUT to 4 channels on an oscilloscope.
1 RT (Resistively Terminated).
2 NT (Not Terminated).
Windows 7 64 bit
Microsoft Internet Explorer 7.0 SP1 or later
Adobe Reader 7.0 or equivalent software for viewing Portable Document Format (PDF) ﬁles
Last Modified: 2015-10-19 00:00:00